set to read AFR from 10-20, so I will use it for sex ads in Durham my example. You will be responsible for paying for your own shipping costs for returning your item. If set to read from 8 22 AFR, the range will be 14, etc. If 30 days have gone by since your purchase, unfortunately we cant offer you a refund or exchange. This is based on HP Tuners. AEMs new X-Series wideband eliminates this sample rate handicap by ditching the serial connection as the primary data source, and instead utilizing an obdii pass-through connection to eliminate unnecessary wires and increase the sample rate ten-fold. Now it will look like this: or for the A/C Pressure switch.
Cheap hookup sites, Wild hookup stories, Sorority hookup vr, Ok hookup review,
Click here for instructions on how to gain access to your PCM. Includes HP Tuners mpvi2 and AEM X-Series Wideband. The formula will be: volts / (5/10) 10 AFR or, (volts /.5) 10 AFR, since HP Tuners Scanner will be using the EGR or AC Pressure switch to get its readings from, we need to add that into the formula. HP Tuners, the most common way to get your aftermarket wideband to communicate with your laptop for tuning and logging was through the use of the mpvi Pro interface or a serial-to-USB adapter. This Wideband O2 sensor is currently the only plug and play solution compatible with the HP Tuners mpvi2 unit and VCM scanner. Package the, mPVI2 together with, vCM Editor and. However, AEM has provided obdii address programmability through the buttons on the gauge face if a different address other than the default address is required, allowing users to program ECU Parameter Identification (PID Message Identifier, and Bit Rate. Now the formula reads like this: (PID.2811/.5)10 AFR, the formula for the A/C Pressure switch is the same, you will just have to change the PID. VCM Suite, and you have one of the most powerful, scanning, diagnostic, logging and calibration products on the market.
Sure, this serial connection may have gotten the data onto your computer, but only after having to manually set up the voltage offset table for your specific sensor and controller, then having to live with data coming in at a much slower sample rate than most tuners really need. Voltage range typically 5v (0-5v some people use the narrowband 02 input (not recommended) which is 1v (0-1v).